Add : 1/F, No.2 Building, 287 Longxi Road, Liwan District, Guangzhou, Guangdong, China
- In 25 Working Days
Spring Hammer Calibration Device
For testing and calibrating spring operated impact hammer, in accordance with IEC60068-2-75/1997-08 Annex B, Figure B.1 to B.4 with scale for impact energies from 0,2J to 1,2J.
The principle of this calibration procedure is to compare the energy provided by a spring hammer, which is difficult to measure directly, to the energy of a pendulum, calculated from its mass and height of fall.
1 Pendulum with rigidly clamped spring at the lower end,
1 frame with bearing for the pendulum,
1 scale plate with drag pointer, scale of anodized aluminum,
2 release bases at the frame,
1 base plate with removable release device.
Impact energy: Max. 2J.
Diameter pendulum: 20mm.
Length pendulum 635mm ± 2.5mm.
Pendulum H : 120mm ± 2mm.
Pendulum D: 20mm ± 0.5mm.
Pendulum W: 4mm ± 0.01mm.
Release device: Ac. Mechanic.
Previous: Vertical Hammer Test DeviceRelated Products
ZLT-CJ1, Single Model ,test the mechanical integri...
ZLT-CJ6, It is for vitreous and porcelain enamels ...
ZLT-BC3, To test the mechanical strength with...
ZLT-SZ1, Conforms to IEC60068-2-75 Test Ehc, ...
ZLT-BC1, To test the mechanical strength with impa...
For testing the mechanical strength and the impact...
ZLT-SE, conform to IEC60068-2-75 table 1 and table...
For testing and gauging the scale of spring hammer...
ZLT-GQ, To determine the mechanical strength ...
ZLT-CD2, For testing and calibrating spring-operat...
ZLT-CJ5, Impact Energy 5J ,test the mechanical int...