Add : 3-7, 1 Floor, Block C, No.23 Panlong Industrial Zone, Longxi Road, Guangzhou, China
- In 25 Working Days
- T/T , D/P
Lateral Pulling Test Apparatus
Product details: ZLT-SP2P
To determine the mechanical strength of the connector, accordance with IEC60320-1:2018 figure 8 and clause 23.3, a lateral pull force is applled in steps of 900±50 as shown in IEC 60320 figure 8a),b),C),d) and e) parallel with
the plane containing the axes of the current-carrying pins/contacts and parallel with the engagement face of the
A pull force according to IEC 60320 Table 11,is applied 50 times in each direction to the cord for 1S±0,5S
1 mounting plated,
1 set of weights,6N-15N-35N, for pull apply,
1 machine device with dimension 40 mm±2 mm.
1 A.C.geared motor with additional gear reduction to approx.60rpm,
Power Supply: A.C.220V 50Hz, other voltages and frequencies on request.
Previous: Test CylinderRelated Products
ZLT-CS1, to test plugs and socket-outlets, cable c...
ZLT-LJ2, for measuring and testing the torque stra...
ZLT-DW2, for continuous bending tests by swivellin...
ZLT-HZ1, installed into the test machine ZLT-DW2, ...
ZLT-LJ3, this device is for testing the resistance...
ZLT-QY1, to determine the dimensional stability of...
ZLT-GT1, is to determine the mechanical strength o...
ZLT-DY1, for testing cable connections with interm...
ZLT-RS1, to determine the verification of resistan...
ZLT-CX1, to determine the fixation of pins in the ...
ZLT-CB1, for verification of the maximum and minim...
ZLT-HP1, for pressure tests on plug pin insulating...